Storage Water Heater Appliance Performance Test Lab With 6 Stations
IEC 60379 Methods for measuring the performance of electric storage
water-heaters for household purposes.
GB 21519 Minimum allowable values of energy efficiency and energy
efficiency grades for electrical storage water heaters.
Equipment system composition:
Test room: The test room has strict temperature and humidity
requirements and is realized by computer room air conditioner.
Sample mounting bracket: 2pcs, installed in the test room, arranged
on both sides, each test bracket has 3 test stations.
Water supply room: include 2 water storage tanks, and constant
temperature and constant pressure water supply line pipe.
Control cabinet: It is the data collection, data processing and
control center of the system. Appliance Performance Test Lab
Mounting of the water-heater:
Wall-mounted water-heaters are mounted on a panel situated at least
150 mm from any structural wall.
They are positioned so that there is a clear space of at least 250
mm above and below the heater and at least 700 mm at the sides and
Floor-mounted water-heaters are placed on the floor or on any stand
supplied with them. A false floor may be used to facilitate
Water-heaters for building-in are built in according to the
|Test room size||6000*4000*2600mm|
|Test room temperature||20 ± 2 °C|
|Test room humidity||<85%|
|Sample mounting bracket||2PCS|
|Test station||6 stations|
|Water temperature range||10-65 °C|
|Water supply line pipe||2 channels|
|Water pressure range||0.1-0.5Mpa. Appliance Performance Test Lab|